|Max Measuring Speed||1.5M/Sec, 60IN/SEC|
|Measuring System||Linear Capacitive Measuring System|
One Alkaline Lithium Button Cell Battery 1.5V
,Type LR44 Capacity-180MAH
|Working Temperature||5℃-40℃/41 to 104 Degree F|
|Note:||Accuracy affected in environements exceeding 80% relative humidity|
1)Operation: Slider can be removed only after the locking screw(#10) is loosened
2)Basic Measuring Methods
a: loosen the locking screw
b: press "on" button, select sae or metric measurement bypressing "MM/INCH" Button.
pressing "MM/INCH" button after measurement will display the sae/metric conversion
c: to reset display to zero, close external measuring jaws,then press"zero" button.
Keep the cailiper clean and dry(liquid can damage the slider)
use a clean, dry cloth to clean all surfaces-do not use solvents or slvent based cleaners
to save power , turn off the unit when not in use
avoid contact with electrocy and or electronic devices
Flashing display indicates low battery. To replace battery, remove battery cover by sliding cover in the direction indicated and replace battery with positive side facing out.
|Display Flashes||Low battery||Replace battery|
|Display does not change when slider is moved||system locked||remove battery and replace after 30 seconds to reset system|
|Accuracy less than ±0.001MM||Dirt in the sensor||Remove slider cover and its assembly, clean with compressed air 5KG/CM2|
|No display on LCD|
1.Improper battery position
1.Remove battery cover and check battery position